article review siobhan lonergan some aspects on the sampling efficiency of microbial impaction air samplers
article review siobhan lonergan some aspects on the sampling efficiency of microbial impaction air samplers. franceso romano jan gustén cesare m. joppolo bengt ljungqvist berit reinmüller.2015 the intent of this article was to assess the sampling efficiency of microbial impaction air samplers and to evaluate the influence of external factors on these measurements. air sampling is the process of analysing and detecting and quantities of contaminants in a particular environment. accurate reliable and efficient air monitoring is of critical importance for ensuring product quality and human health. air monitoring requires a particular type of aerosol measurement which monitors the viable airborne particles ranging in size from 0.001 to 100 ?m 1 these viable airborne particles are collected using an air sampler fitted with an appropriate culture media such as tryptone soya agar which acts as a nutrient source for a wide range of aerobic micro-organisms. this media is incubated for a defined time period and at a defined temperature. resultant colonies are enumerated and identified to species level. results are expressed as colony forming units per cubic meter of air sampled expressed cfu/m3 even though air samplers rely on the same basic technology the results from different air samplers are not comparable. therefore the sampling performance must depend on other external factors such as microbial load environmental conditions bioaersol source and specific instrument features 1 8 and 11 the absence of an industry accepted standard procedure for sampling air monitors is also an influential factor. considering all these facts it is essential to choose an appropriate air sampler and of equal importance is how to interpret its results. ljungqvist & reinmuller demonstrated that air samples were not comparable with a tenfold difference in cfus between some samplers 6 air classifications for sterile products by the fda who and euggmp 3 10 &2 are < 1 cfu/m3 which translates as zero colonies detected per m3.consequently with air samplers not being comparable it is possible to have different microbial results from different air samplers that have sampled the same environment 4 11 thus an oos environment may appear in specification. over the years it has become apparent that the behaviour of airborne particles is determined by characteristics such as their size shape and density. the aerodynamic flow around the particle depends on particle size which is measured in terms of the particles aerodynamic diameter. the most commonly used metrics when describing particle size distributions are d values. d-value can be thought of as a mass division diameter. it is the diameter which when all particles in a sample are arranged in order of ascending mass divides the samples mass into specified percentages. the percentage mass below the diameter of interest is the number expressed after the d. the d50 is the diameter of the particle that 50% of a samples mass is smaller than and 50% of a samples mass is larger than. the d50 is also known as the mass median diameter as it divides the sample equally by mass 5 this study aimed to assess the performance of four different microbial air samplers when exposed to different microbial bioburdens generated by human activity. the microbial bioburden will be greatly influenced by cleanroom garments and the behaviour of the operator. in addition different quality clothing systems reduce air contamination generated by humans to different degrees 7 9 a controlled experiment was set up in a cleanroom environment where three types of cleanroom garments were chosen deemed to be of low medium and high contamination levels and the operator was instructed to walk in front of the air monitor for a time period of ten minutes. the experiment was repeated in triplicate using a standard operating procedure. the three tests clearly showed that movement and clothing had a major effect on contamination levels in a cleanroom environment. a direct correlation between the highest quality cleanroom garment with the lowest level of microbial bioburden of cfus/m3 was established. it was also observed under identical environmental conditions two air samplers with varying d50 values displayed a marked difference in the accuracy of the result. the sampler with the lower d50 value more accurately monitored variations in microbial concentration. in addition the efficiency of the air sampler to detect small particles is important. it was ascertained that more than half of the viable particles were smaller than 3.64 ?m. this indicates high efficiency samplers with low d50 values should be employed for air monitoring in cleanrooms. i strongly agree with the outcome of this study which confirms that humans their movement and the quality of cleanroom garments have a major influence on particle counts in the cleanroom environment as people are a huge source of contamination in fact the biggest source of viable and non –viable contamination this was collaborated by hu s and a.shiue study relating to the validation and application of the personnel factor for the garment used in cleanrooms 13 their studies proved that human movement and the aging of cleanroom garments contributed to an increase in microbial load. each adult loses about 6-14 grams of dead skin material every day. each person loses a complete layer of skin about every four days equivalent to 10 000 000 particles per day. ordinary walking movements emit about 10 000 particles per minute. hence this article proved if the cleanroom garment is of low grade fabric these particles will be released to the air and act as carriers for microorganisms. in the pharmaceutical industry and especially in sterile manufacture it is imperative that quality know if any airborne contaminants put products at risk. it is also important to understand what the contaminants are and their origin. critical environmental measurements like airborne particle measurements assist with keeping processes in control and with helping to find the root cause of unwanted events. air samplers are an important tool in the armoury of the microbiologist to monitor cleanrooms for particles and other potential contaminants. successful air monitoring requires awareness understanding and action. awareness of what the data is saying what counts are present and where and what organisms are present are where. whats measured must be managed. knowledge without action is useless. air monitoring results ultimately need to drive you to take actions if required. therefore the theory which this article proves that though air samplers rely on the same basic technology the results from different air samplers are not equivalent. it clearly demonstrates the physical parameters of the air monitor such as the d50 as well as the concentration of microorganisms and environmental conditions have a major bearing on the results obtained. this article has led to a better understanding of air sampler performance and the importance of impaction theory. in summation this study confirms the importance of a standard methodology for microbial air monitoring and that understanding the importance of the dynamics of air sampling in relation to collection of particles is paramount. results between different air samplers are not comparable and therefore an air sampler should be assessed and selected on sampling requirements sampling conditions and the characteristics of the air sampler. the study revealed the microbial load is influenced by operator garments and behavior. the utilization of high quality cleanroom clothing can reduce microbial bioburden. results also indicated that d50 value of a sampler can aid in predicting its performance and assist in interpreting results. air monitoring programs are integral components of a process to monitor microbial levels in a cleanroom and to ensure the safety efficacy and quality of bio-susceptible products. air monitoring can detect changes in the contamination recovery rate that may be indicative of changes in the state-of control within the environment. this is to provide confidence that routine air monitoring will mitigate against microbial proliferation and product contamination. references baron p. a. & willeke k. 2001 aerosol measurement principles techniques andapplications 2nd ed. new york: john wiley & sons. eudralex. the rules governing medicinal products in the european union volume 4 eu guidelines to good manufacturing practice medicinal products for human and veterinary use annex 1 manufacture of sterile medicinal products: euggmp 2008 recommended limits for microbial contamination. fda. sterile drug products current good manufacturing practice air classifications by usfda guideline on sterile drug products frankel m. timm m. hansen e. w. & madsen a. m. 2012 comparison of samp-ling methods for the assessment of indoor microbial exposure. indoor air 22 405–414. internationalorganization for standardization iso 13317-1-2001 determination of particle size distribution by gravitational liquid sedimentation methods part 1: general principles and guidelines ljungqvist b. & reinmüller b. 1998 active sampling of airborne viable parti-cles in controlled environments: a comparative study of common instruments.european journal of parenteral sciences 3 59–62. .ljungqvist b. reinmüller b. & tammelin a. 2012 comparison between two mea-suring methods with active air sampling of airborne viable particles. europeanjournal of parenteral sciences 17 56–59.. seinfeld j. h. 1986 atmospheric chemistry and physics of air pollution. new york:john wiley & sons. tammelin a. ljungqvist b. & reinmüller b. 2013 single use of surgical cloth-ing system for reduction of airborne bacteria in the operating room. journal ofhospital infection 84 245–247. who 2002: air classifications as per who 2002 willeke k. 1976 temperature dependence of particle slip in a gaseous medium.journal of aerosol science 7 381–387. zhen s. li k. yin l. yao m. zhang h. chen l. et al. 2009 a comparison of theefficiencies of a portable biostage impactor and reuter centrifugal sampler rcs high flow for measuring airborne bacteria and fungi concentrations. journal ofaerosol science 40 503–513. hu s shiue a. validation and application of the personnel factor for the garment used in cleanrooms. building and environment. 2016;97:88-95.
Or Upload Your File: .pdf, .docx, .txt.
Top of Form
Drop Your File Here or Click to Upload
Created with Sketch.
Bottom of Form
Check My Essay
By clicking Check Topics, Sample Papers & Articles Online for Free, you agree to our TermsArticle Review.docx
Browse new file